Verify that the device beeps when therapy is stopped. The user of this device should make sure it is used in such an environment. From the Monitor Pressure screen, press and hold the Wheel for approximately 2 seconds to turn off therapy and return to the Main Menu. Monitor Pressure Screen From the Main Menu, if you select Therapy and then press the Wheel, the following Monitor Pressure screen appears. Press the tab in the hole on top of the tank in toward the front of the tank.
Empty any remaining water from the base of the tank. Press in the tab on the side cover and pull the cover away from the unit. You can also use the following address: Respironics Inc. This button lights up when therapy is active. If the problem there is significant air adjustment or improper continues, contact your home care leakage around the mask fitting.
The user of this device should make sure it is used in such an environment. If using the humidifier with the device, check the humidifier settings. Refer to the humidifier instructions to make sure the humidifier is working properly. Repairs and adjustments must be performed by Philips Respironics- authorized service personnel only. You can also use the following addresses: Respironics, Inc.
. Exhaled Tidal The estimated Exhaled Tidal Volume is obtained by the integration of patient flow. Replace any damaged parts before using. Electromagnetic propagation is affected by absorption and reflection from structures, objects, and people. Unauthorized service could cause injury, invalidate the warranty, or result in costly device damage. A properly installed, undamaged It must be in place at all times when the device is operating.
Refer to the instructions provided with the accessory for more information. If the foam filter is torn, replace it. Discard and replace any damaged parts. Note 2: These guidelines may not apply in all situations. A setting of 1 provides a small amount of pressure relief, with higher numbers providing additional relief. The figure below illustrates some of the device features, described in the table below. Only Philips Respironics-supplied filters should Never install a wet filter into the device.
Reassemble the tank by placing the hinges on the tank lid over the 2 tabs on the back of the tank base. A mask should not be used unless the device is turned on and operating properly. Device Alarms This chapter describes the ventilator alarms and what you should do Note: If multiple alarms occur at the same time, only the if an alarm occurs. Contact Ventilator Inoperative your home care provider for service. Note: For information on troubleshooting your therapy device, see the manual included with your therapy device. The settings are 0 which turns the heater plate off , 1, 2, 3, 4, and 5. Device Setup Installing the Air Filter The device uses a gray foam filter that is washable and reusable.
Volume Vte The display is updated at the end of each breath. Spontaneous Pressure Support; A Bi-level therapy mode where breaths are patient-triggered and patient-cycled. The data may be corrupted. You must ensure sufficient drying time for the cleaned filter. Rise time cannot be adjusted when Bi-Flex is enabled. If you need to contact Respironics directly, call the Respironics Customer Service department at 1-724-387-4000 or +49 8152 93060. See the Troubleshooting section later in the manual for additional information.
If the device does not operate accordingly, contact your home care provider, as the alarm system may not be fully functional. One gray foam filter is required for reusable gray foam filter is supplied with your device. Hinges Tank Lid Middle Seal. Contact your home care provider with any - or - if Icon option is questions. If this is not comfortable, you can increase or decrease the setting from 1 to 3. Cleaning Do not immerse the device in liquid or allow any liquid to enter the enclosure or inlet filter. Contraindications The device is contraindicated on patients without a spontaneous respiratory drive.
Plug the socket end of the power cord included into the power supply also included. Close the lid until the tab on the lid snaps back under the lip in the tank base. Dishwashing can be performed once a week. The user of this device should make sure it is used in such an environment. This selection allows you to control the setting must be set to 15.